Measure SYN RF Cavity Voltage using X-ray end point energy technique
Measure SYN tune vs RF cavity voltage until 17:20
X-ray measurement vs RF cavity voltage and SYN tune measurement
-Modify timing and pattern memory
16:15 - Change everything back to normal operation
-Do not forget to plug in the Trigger signal for DG-535 No#2, this is for Kicker and STR Septum.
I would like to announce that the new electronic log book for Accelerator Technology Division has been installed.
Beam @50 mA
Warm Up HBT, STR Magnets & Flush TGP 1 Cycle
Stop Injection @ 135 mA
Ramping complete @ 133 mA
Before - dRMSx 0.0741 mm
dRMSy 0.1330 mm
After - dRMSx 0.0194 mm
dRMSy 0.0239 mm
Dump Beam @ 50.5 mA
Warm up magnets HBT,STR magnets & Flush TGP 1 cycle
Stop Injection @ 131.8 mA
Ramping Complete @ 130.9 mA
Before dRMSx 0.0662 mm.
dRMSy 0.1368 mm.
After dRMSx 0.0186 mm.
dRMSy 0.0232 mm.
Beam Drop 55mA to 52mA
Dump Beam @ 46mA
Warm Up HBT, STR magneta & Fluah TGP 1 Cycle
RF-SYN sometime NOT trig (often)
Stop Injection @ 134mA
Ramping complete @ 133mA
Before - dRMSx 0.0675 mm
dRMSy 0.1239 mm
After - dRMSx 0.0223 mm
dRMSy 0.0229 mm
By re-initialize LBT and SYN magnets.
SYN-RF is not working sometime. (Trigger in control room OK, but SYN-RF PS no power)
Need for checking !!!
Dump Beam @ 48 mA